5 pg/ml - 2000 pg/ml
Recommended Dilution (serum & plasma):
This ELISA kit shows no cross-reactivity with any of the cytokines tested: Human Angiogenin, BDNF, BLC, ENA-78, FGF-4, IL-1 alpha, IL-1 beta, IL-2, IL-3, IL-4, IL-5, IL-7, IL-8, IL-9, IL-10, IL-11, IL-12 p70, IL-12 p40, IL-13, IL-15, I-309, IP-10, G-CSF, GM-CSF, IFN-gamma, Leptin, MCP-1, MCP-2, MCP-3, MDC, MIP-1 alpha, MIP-1 beta, MIP-1 delta, PARC, PDGF, RANTES, SCF, TARC, TGF-beta, TIMP-1, TIMP-2, TNF-alpha, TNF-beta, TPO, VEGF.
Number of Targets Detected:
Compatible Sample Types:
- Cell Culture Supernatants
Method of Detection:
1, 2, or 5 x 96-Well Strip Microplate Kit
Protein Name & Synonyms:
Macrophage colony-stimulating factor 1 (CSF-1) (M-CSF) (MCSF) (Lanimostim) [Cleaved into: Processed macrophage colony-stimulating factor 1]
The RayBiotech human M-CSF ELISA employs an antibody specific for human MCSF coated on a 96-well plate. Standards and samples are pipetted into the wells and MCSF present in a sample is bound to the wells by the immobilized antibody. The wells are washed and biotinylated antihuman MCSF antibody is added. After washing away unbound biotinylated antibody, HRPconjugated streptavidin is pipetted to the wells. Thewells are again washed, a TMB substrate solution is added to the wells and color develops in proportion to the amount of MCSF bound. The Stop Solution changes the color from blue to yellow, and the intensity of the color is measured at 450 nm.
The minimum detectable dose of human M-CSF was determined to be 5 pg/ml. Minimum detectable dose is defined as the analyte concentration resulting in an absorbance that is 2 standard deviations higher than that of the blank (diluent buffer).
- Strip plates and additional reagents allow for use in multiple experiments
- Quantitative protein detection
- Establishes normal range
- The best products for confirmation of antibody array data
Recovery was determined by spiking various levels of Human M-CSF into the sample types listed below. Mean recoveries are as follows:
- Pre-Coated 96-well Strip Microplate
- Wash Buffer
- Stop Solution
- Assay Diluent(s)
- Lyophilized Standard
- Biotinylated Detection Antibody
- Streptavidin-Conjugated HRP
- TMB One-Step Substrate
Other Materials Required
- Distilled or deionized water
- Precision pipettes to deliver 2 µl to 1 µl volumes
- Adjustable 1-25 µl pipettes for reagent preparation
- 100 µl and 1 liter graduated cylinders
- Tubes to prepare standard and sample dilutions
- Absorbent paper
- Microplate reader capable of measuring absorbance at 450nm
- Log-log graph paper or computer and software for ELISA data analysis
- Prepare all reagents, samples and standards as instructed in the manual.
- Add 100 µl of standard or sample to each well.
- Incubate 2.5 h at RT or O/N at 4°C.
- Add 100 µl of prepared biotin antibody to each well.
- Incubate 1 h at RT.
- Add 100 µl of prepared Streptavidin solution to each well.
- Incubate 45 min at RT.
- Add 100 µl of TMB One-Step Substrate Reagent to each well.
- Incubate 30 min at RT.
- Add 50 µl of Stop Solution to each well.
- Read at 450 nm immediately.
Need your results faster? Try RayBiotech's SpeedELISA platform for quantitative detection in just three hours.
The entire kit may be stored at -20°C for up to 1 year from the date of shipment. Avoid repeated freeze-thaw cycles. The kit may be stored at 4°C for up to 6 months. For extended storage, it is recommended to store at -80°C.